V-Beam Pulsed Dye Laser
Candela V-Beam Laser, 595nm (Pulsed-Dye Laser)
Total Shots: 101,000
Includes: 3 pair eyewear, 7/10/3x10mm handpieces, Distance Guages, Operators Manual, Patient Video and System Key.
System will need a New Dye-Kit installed by qualified service technician prior to operation.
V-beam has maintained Gold Standard with 15+ years of use and 300+ published papers, the V-beam delivers the safety and efficacy of the clinically proven pulsed dye laser.
REQUEST MORE INFORMATION
Candela V-Beam System Specifications:
Laser Type:Flashlamp-excited, pulsed dye laser
Wavelength:595 nanometers (nm)
Optical Output:Lens-coupled optical fiber with user selectable slider spot sizes.
Maximum Energy:6 joules (J)
Stability of Output Energy:±14%
Pulse Repetition Rate:1.5 Hz repetitive pulsing
Pulse Duration:0.45-40 milliseconds
Beam Spot Size:5 mm, 7 mm, 10 mm, 3×10 mm
Cooling Method:Ambient air
Dimensions:(HxWxD) 43.5x 18x 32 (111 x 46 x 81 cm)
Weight:300 lbs (140 kg)
System will require Installation & Calibration of Dye Kit to be Patient Ready.
V-Beams versatility enables you to address a wide range of vascular lesions, including: periorbital wrinkles, facial veins, rosacea, leg veins, port wine stains, hemangiomas, scars, stretch marks and warts.
V-beam has maintained Gold Standard with 15+ years of use and 300+ published papers, the Vbeam delivers the safety and efficacy of the clinically proven pulsed dye laser. V-beam’s versatility enables you to address a wide range of vascular lesions, including: periorbital wrinkles, facial veins, rosacea, leg veins, port wine stains, hemangiomas, scars, stretch marks and warts. V-beam offers DCD cooling for effective cooling and protection of the epidermis offering your patients greater comfort.
On-Line Ordering Not Available / Must Contact to Request Purchase Details.
FDA requires all listings in this category to include the following text: “The sale of this item may be subject to regulation by the U.S. Food and Drug Administration and state and local regulatory agencies. If so, do not bid on this item unless you are an authorized purchaser. If the item is subject to FDA regulation, we will verify your status as an authorized purchaser of this item before shipping of the item.”
DISCLAIMER: Regardless of the origin of the equipment, documentation provided or identification appearing upon the equipment, the equipment described and offered here is in no way certified for, recommended for, or offered for any specific use. The purchaser agrees that the seller shall not be held responsible or liable for any injuries or damages, whether incidental or consequential, associated in any way with the equipment. The purchaser, by bidding on this equipment, indicates their acknowledgment of, and agreement to the terms of this disclaimer.